High-temperature sample heating stage with strip heater for powder diffraction. Allows for very fast heating and cooling and ensures high sample position stability with heating strip pre-stressing.
with Pt-strip: 25 °C to 1600 °C in air, vacuum
with Ta and C-strip: 25 °C to 1500 °C in vacuum
Atmospheres: air, inert gas, vacuum(10⁻⁴ mbar)